E5071C MANUAL PDF
The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||12 October 2006|
|PDF File Size:||19.92 Mb|
|ePub File Size:||9.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
There are two procedures in accordance with the HDD revision.
This calibration effectively eliminates the directivity error, crosstalk, source match error, frequency response reflection tracking error, and frequency response transmission tracking error from the test setup in a transmission or reflection test using those ports.
EC firmware is executed, then the measurement display appears. For more information, see Modifying Calibration Kit Definition.
Operation of any electrical instrument in such an environment clearly constitutes a safety hazard. Inset the key in the lock and turn anti-clockwise.
In this case, follow the instruction to return to the previous registration screen. Click Manula to start the calibration measurement. Remove the hard disk.
The procedure consists of the following items. Factory recovery Returns the contents of the C drive to the factory state. Touch the x marks in that order with your finger.
EC User Manual – EC Network Analyzer from Keysight
Figure Recover Factory Backup ecse Step 8. The HDD revision can be checked using the following procedure: Select the D drive to restore, then click Next button.
Agilent Technologies assumes no liability for the customer’s failure to comply with these precautions. Safety Symbols Instruction Manual symbol: The firmware revision you installed mqnual even if you perform factory recovery. It’s a community-based project which helps to repair anything.
Help File – User Manual for E5071C ENA Network Analyzer (Operation and Programming)
If you perform the following procedure incorrectly, a message asking you whether to return to the previous registration screen and perform the registration appears. To use this function, you must prepare for recovery in advance. Click Done to terminate the full 2-port calibration process.
The dialog box disappears automatically after a mahual seconds. The following screen appears during backup. Turn the screw in anti-clockwise direction until it is locked. A warning message is displayed. A total of twelve error terms, six each in the forward direction and the reverse direction, are used in a calibration. Select a test port and corresponding S parameter on which 1-port calibration will be performed.
Turn on the EC.
Agilent Technologies EC Instruction manual |
Insert the hard disk into the slot. The Confirm Device Removal message box appears. Turn the key to the left until it is locked. Connect the keyboard to the EC. Figure 1 Item 1: Turn off the power.
A chassis terminal; a connection to the instrument’s chassis, which includes all exposed metal structure.