FLUKE 80T-IR PDF
80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
|Published (Last):||19 April 2006|
|PDF File Size:||11.53 Mb|
|ePub File Size:||10.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
Accuracy figures are valid for a period of one year. Adjust the setpoint of the Blackbody to output each of the temperatures in Table 2 at 0. Save to parts list Save to parts list. Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody plate. Thank you for your feedback.
Fluke Thermometer 80T-IR User Guide |
You have chosen to save the following item to a parts list:. The Manufacturers disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from your use of or your inability to use the Information downloaded from this website.
Power up the UUT and allow it to stabilize for five minutes. The Manufacturers reserve the right to change this Information at any time without notice.
With the UUT powered on, check that the battery voltage is at least 7. The foregoing information relates to product sold on, or after, the date shown below.
The product does flue contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering. Save this item to a new parts list. In between measurements move the UUT to an environment not affected by the heat source. Please enter a message. Add to a parts list.
Fluke 80T-IR Thermometer User Manual
Verification Points Model Blackbody Temperature at 0. Record the reading displayed on the DMM.
You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice. Replace the battery if necessary. Please select an existing parts list.
Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure. If your instrument fails to pass any performance specifications, return it to the Eluke Eactory or Service Center for repair. May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent accuracy, i.
Fluke 80t-ir Infrared Temperature Probe 80tk Thermocouple Module
Sending feedback, please wait Search the history of over billion web pages on the Internet. Refer to Figure 1 for equipment configuration. Reactivate by switching the UUT off and on again. Repeat steps 1 through 5 for the remaining verification points called out in Table 2. Set the DMM to the 3.
The product detailed below complies with the specifications published by RS Components.
Save to an existing parts list Save to a new parts list.